HOW TO FIT THE TAYCO RECOVERX BRACE
Wearing the RecoverX properly is imperative to recovery. Watch the video below to learn how to put on the brace.
How to use the stirrup strap adjustment tool
Slide the tool under the strap, ensuring the two tines go around the peg
Pull the strap to the desired hole setting
Remove the adjustment tool and press the strap down around the peg
Easily change the range of motion on your RecoverX by watching the video below.
Helpful Tips for Hinge Adjustment
- If the locking bars are removed, store them in a place you won’t forget in case you need to re-lock the hinge.
- Ensure the retaining button on the locking bar is pointed away from the brace so that it will interlock with the corresponding hole in the hinge.
- Make sure both hinges are set in the same position for optimal recovery.
If the brace feels too loose, unstable or uncomfortable, use this guide to help diagnose and fix the issue. If you have more questions, please call/text us at 574.855.8213 for live support or email us at firstname.lastname@example.org.
According to a research study conducted by Ball State University, your TayCo Brace provides 96% of the stability of a walking boot for the ankle. However, a small amount of motion is expected when wearing the brace.
In the fixed position, the uprights should fit parallel to the lower leg. If they do not, refit while ensuring your knee is bent slightly more than a right angle as you put on the brace and tighten the straps.
If the straps are too short, please call or email us and we will send you longer straps.
Lack of Stability
If the brace does not feel stable, the biggest culprit will likely be the stirrup strap. If it's not flush with the bottom of your shoe when wearing the brace, it's not tight enough. With your shoe in the brace, start with tightening the stirrup strap one hole at a time until you can't go any further.
If the heel part of the brace moves up and down significantly or rubs against the ground, this may be a sign that the stirrup strap is not tight enough. It could also mean that the brace is not appropriately aligned at the heel where the sole meets the body of the shoe.
If adjusting the stirrup strap does not increase stability, retighten straps 1, 2, 3, 4 in sequence as tight as you can while maintaining comfort.
If you feel discomfort or pinching on the calf or shin, double-check you've centered the brace on your leg, framing equal parts calf and shin.
If you feel pressure on the bony prominence of your ankle, it's because the circular pads don't align with your ankle or your ankle deformity may be too great for an over-the-counter brace. This could also mean the brace is sitting too high or low, or you may also need a different size.
If the footplate is creating too much pressure or is uncomfortable as it extends towards the toes, please contact us.
If the metal ring on straps 3 and 4 is rubbing your leg, reposition the straps to make sure the metal ring aligns with the edge of the upright.
- The TayCo RecoverX Brace is designed for use only on the ankle and outside the shoe.
- Do not use for any other part of the body.
- Inspect the brace before each use. The brace must not be used if any components are torn or broken. Do not attempt to repair.
- Any alteration of application other than illustrated is misuse of the TayCo RecoverX Brace. Use only as instructed.
- Proper use is necessary to avoid risk of further injury
- This product is not a substitute for medical care
- TayCo Brace, Inc. will not be liable for injuries or damages resulting from misuse or misapplication of the TayCo RecoverX Brace
- Discontinue use if you experience irritation, discomfort or instability related to the brace
- Always consult a doctor or medical professional for diagnosis and treatment of any pain or injury, or if pain or discomfort continues or gets worse